Dry heat may be used to sterilize patient-care items that might be damaged by moist heat (e.g., burs and certain orthodontic instruments). Although dry heat has the advantages of low operating cost and being noncorrosive, it is a prolonged process and the high temperatures required are not suitable for the sterilization of many instruments and devices. There are two types of dry-heat sterilizers used in dentistry: static-air and forced-air types.
The static-air type is commonly called an oven-type sterilizer. Heating coils in the bottom or sides of the unit cause hot air to rise inside the chamber through natural convection. The forced-air type is also known as a rapid heat-transfer sterilizer. Heated air is circulated throughout the chamber at a high velocity, which permits more rapid transfer of energy from the air to the instruments, thereby reducing the time needed for sterilization.2,3
Your session is about to expire. Do you want to continue logged in?
WARNING! You did not finish creating your certificate. Please click CONTINUE below to return to your previous page to complete the process. Failure to complete ALL the steps will result in a loss of this test score, and you will not receive credit for this course.